|Place of Origin:||China|
|Minimum Order Quantity:||negotiatable|
|Packaging Details:||1 pc/packing , single ,packed in box then carton|
|Delivery Time:||10-15 working days after received the payment|
|Payment Terms:||T/T, Western Union, MoneyGram|
|Supply Ability:||500 pcs / month|
|Product Name:||Endodontic Wireless ZR-AUTO Motor||Size:||ZR-AUTO|
|Standard:||CE/ISO/FDA/FSC||Application:||Root Canal Treatment|
dental endodontic motor,
rotary endo motor
Endodontic Wireless Muti-function Touch-Screen Endo Motor ZR-AUTO Motor
1. The device brief summary:
ZR AUTO MOTOR” is a compact device combining endodontic motor with microprocessor control for working with rotating Ni-Ti instruments.
2. Field of application:
The device is designed for use in dentistry (endodontics) and can be used only in hospitals by medical specialists licensed to practice dentistry.
The manufacturer is not responsible for the device misuse.
Table 1 The list of programs endomotor
|The name of the file system, under which adapted the program|
1 ,2, 3, 4
|forward 170° and reverse 50º|
|reverse 150º and forward 30°|
|8||reverse 170º and forward 50°|
Continuous forward rotation normally, but when the load on the file exceeds the set limit, the file automatically starts alternately between forward 180° and 90º reverse rotation
Continuous” rotation mode - instrument constantly rotates clockwise or counterclockwise, respectively
“Reciprocal” rotation mode - instrument alternately rotates clockwise and counterclockwise with clearly fixed speed and rotation angle in both directions: with a big rotation angle when moving forward (clockwise or counterclockwise, respectively) and with a small when moving back.
4. Safety measures and warnings
part 3. “Accessories”).
case of the device integrity damage warranty is considered to be invalid. The device power source should be changed by authorized service departments only.
in the presence of inflammable anesthetic mixtures with air, oxygen or nitrogen oxides.
sterilized and disinfected directly before the first device use and also after each patient in order to avoid cross infection (for more details see part7 “Cleaning, Disinfection and Sterilization”).
previous patient. Avoid lubricant ingress into the micromotor expanded parts.
vibrations), stop using the device immediately and try to find the cause of malfunction, basing on the information represented in the part9.
.Carefully follow manufacturer’s instructions con- cerning use of the endodontic instruments and avoid use of worn-out and damaged files.
such a file may result in injury.
not more than 80%, atmosphere pressure (101±3) kPa. Any violation of the pointed restrictions may lead to the device misoperation.
• In cases where a patient has been fitted with an implanted heart pacemaker (or other electrical equipment) and has been cautioned against the use of small electrical appliances (such as electric shavers, hair dryers, etc.) it is recommended not to use the ZR AUTO MOTOR Plus
• Do not use .ZR AUTO MOTOR for implant or any other dentist procedure outside endodontics.
There are no known adverse reactions..
7. Legal Manufacturer
SHENZHEN FEIHUAN MEDICAL INSTRUMENTS CO., LTD
8. EU Representative
Well Kang Limited
The Black Church, St.Mary's Place, Dublin 7, Ireland
|Dental Root Canal Treatment||Hospital , Clinic , lab...|